Before using this unit for the first time, please be sure to read this instruction manual carefully and use the unit properly. Know Your Unit Inserting and Replacing Batteries Setting the Date and Time
|Published (Last):||5 May 2016|
|PDF File Size:||17.15 Mb|
|ePub File Size:||12.11 Mb|
|Price:||Free* [*Free Regsitration Required]|
Before using this unit for the first time, please be sure to read this instruction manual carefully and use the unit properly. Know Your Unit Inserting and Replacing Batteries Setting the Date and Time Each ECG reading records about second measurement and these ECG readings can help your doctor monitor your condition.
Never use the unit to monitor vital signs and acute events. The measurement results are for monitoring purposes only, and should never be used as a basis for starting or modifying treatment without independent confirmation by medical examination. Omron does not recommend the use of contact gel. Selects the highlighted menu option or ECG data. Chest electrode Clears error messages.
Page 7 HCGE. SD memory card M. Storage pouch Page 8 HCGE. Page 9 HCGE. Calibration guide V. SD memory card indicator Inserting and Replacing Batteries Slide the battery cover in the direction of the arrow, and lift the end of the cover upwards.
Note: Do not pull too hard on the cover. Page 12 HCGE. Setting the Date and Time Always set the date and time before using the unit for the first time. Check the date and time are correct before using the unit, reset them if necessary. Page 14 HCGE. The day setting is set and the month setting is highlighted.
Page 16 HCGE. Page 17 HCGE. Page 18 HCGE. Only eject an SD memory card when the unit is turned off. Using SD Memory Card 4. Page 20 HCGE. Format SD? Before taking a measurement for the first time, read these instructions carefully and make sure you observe the following points each time you take a measurement. Taking a Measurement Press the button to turn the unit on. The unit displays a reminder that self-measurement is not the same as medical diagnosis and then the guidance screens are displayed.
The guidance screens are displayed twice. Page 23 HCGE. Taking a Measurement Place the chest electrode on bare skin about 5 cm below your left nipple. Page 24 HCGE. Measurement ends Measurement starts During measurement The unit beeps every second. About 30 seconds The measurement ends automatically, indicated by 4 rapid beeps. Page 25 HCGE. Taking a Measurement A measurement result message is displayed when the measurement has been completed. Refer to p. If you press the OK button while the result message is displayed, the list of ECG data stored on the SD memory card or unit is displayed.
Page 26 HCGE. Taking a Measurement Common Causes of Inaccurate Measurement Results The following situations can result in inaccurate measurement results or errors.
Page 27 HCGE. Taking a Measurement Measurement Result There are 13 messages regarding the measurement result. A letter of the alphabet indicates each of the 13 possible messages. This letter is displayed together with the time and date of the measurement when the GRAPH menu is displayed. Page 29 HCGE. The most recent measurement result is displayed at the top of the list and is automatically highlighted, with any previous measurement results listed below it in chronological order.
Page 31 HCGE. Adjusting the Settings Press the button to select the desired setting, then press the OK button to see the options for that setting.
Adjust the settings then press the OK button to confirm the changes. Adjusting the Settings 8. For details on setting the date and time, refer to p.
ECG Monitor HCG-801
Omron HCG-801 Manuals
The revolution in blood pressure monitoring
Omron HCG-801 Instruction Manual